Lower-speed Centrifugal Machine is mainly used to separate suspension of solid particles and liquid; Or emulsion in the two kinds of different density and mutual miscibility of liquid separation (such as isolated from milk cream); It can also be used to eliminate wet solid in the liquid, such as use washing machine to jilt dry wet clothes; Special speed tubular can separate different density of gas mixture, such as enrichment and separation of gaseous uranium hexafluoride: With different density and particle size of solid particles in a liquid sedimentation velocity of different characteristics, some sedimentation centrifuge for solid particles according to density and grain size grading.
The table-centrifugal machines are widely used in the laboratory or produce department fororganisms? medical? chemistry etc. The lower speed centrifugal machine with the maximum speed 4000r/min and be equipped with/without timer. It mainly be used for the appraisal radioactivity and separate cell or particle.
Put the machine on a level and strong table and keep the working room clear ?dry and ventilated .Indoor temperture5- 32? is ok Open the power supply and make sure the indicator lamp is light .Shut the power switch. Open the cover of the machine, clear the rotor and container. Put the test tube (with same weight) into the machine .put down the cover of the machine .Adjust the timer as your requirement. Open the power switch. Adjust the governor as you requirement. Before you are going to stop the work, you should adjust the governor to "0"fist, and then shut the power supply .Clean the container and rotor, then put down the cover. .
Time rang:0~60min or without limit
Electric current:AC110V 60 HZ
Maximum relative Centrifugal force: 1435×g